Generate a pair of quadrature-phase digital signals.

*Stefano Salvatori and Marco Girolami*

The circuit in this Design Idea realizes a simple, low-cost lock-in amplifier employing an Analog Devices AD630 balanced modulator-demodulator IC (Reference 1). The device uses laser-trimmed thin-film resistors, yielding accuracy and stability and, thus, a flexible commutation architecture. It finds use in sophisticated signal-processing applications, including synchronous detection. The amplifier can detect a weak ac signal even in the presence of noise sources of much greater amplitude when you know the signal’s frequency and phase.

As an analog multiplier, the AD630 reveals the component of the input-voltage signal in a narrow band around the frequency of the reference signal. The lowpass filter at the AD630’s output allows you to gain information on the weak signal amplitude, which the uncorrelated noise originally masked. When the input voltage and the reference voltage are in phase, the lowpass filter’s output, V_{OUT}, assumes the maximum amplitude. Conversely, if the input voltage and the reference voltage are in quadrature, the output voltage would ideally be 0V. In this way, if both in-phase and quadrature reference signals are available, two balanced demodulators reveal the in-phase output voltage to be 0° and the in-quadrature output voltage to be 90°. You can calculate the module and phase shift as follows:

The two AD630s have a gain of ±2 and receive the amplified signal, V_{IN}, through two identical amplifiers, A_{1} and A_{2}. At Pin 7 of IC_{1}, a bipolar ±5V squared signal appears in phase with the reference signal. OA_{1} integrates the amplifier voltage, which generates a triangular wave that IC_{2}’s comparator compares with the V_{R2} voltage. You must regulate V_{R1} and V_{R2} to obtain a perfect 90°-shifted command for IC_{2}. You can monitor the voltage at IC_{2}’s Pin 7. Measurement accuracy and repeatability depend strongly on the RC time constant of the integrator and the values of V_{R1} and V_{R2}.

Figure 1. OA_{1} integrates the bipolar V_{A} signal and creates a triangular wave. V_{R1} and V_{R2} obtain a 90°-shifted reference voltage with respect to V_{A}. |

You can use a different approach to generate in-phase and in-quadrature reference signals. Figure 2 shows an all-digital circuit, which you can implement in a small CPLD (complex programmable-logic device) to generate the 0 and 90° reference signals in Figure 1. Counter 1 measures the reference-signal time in terms of the N number of digital clock pulses, where the reference time can be different from 50%. It receives a preset command at the N_{1}=1 value at each positive front edge of the reference signal. D-type flip-flop IC_{1} generates such pulses. At each positive edge of the reference signal, IC_{2} acquires the N/4 value. Meanwhile, Counter 2 counts the clock periods and receives a restart command at the N_{2}=1 value when its value reaches the comparator-measured N/4 quantity.

Figure 2. You can implement this all-digital circuit in a small CPLD |

To overcome the lack of the last EQ signal when the reference time is greater than approximately four times the N/4 integer value, the OR combination of the two RST and EQ pulses yields four almost-equidistant positive-edge commands in each reference-time period. The N/4 integer division, a logical right shift by 2 bits of N

To maintain accuracy at least comparable with that of the AD630, the N_{1} output of Counter 1 would be the highest. However, an increase in the number of bits decreases the maximum reference frequency for a given digital-clock frequency if you want N_{1} to reach high values. For example, if N is 15 bits, the N_{1} output assumes the 32,767 maximum value with an accuracy of approximately 0.01%. If the reference-time period decreases, you can assume a minimum value of 3277—that is, one-tenth of the maximum value—for N_{1}, with a correspondingly lower accuracy of 0.1%, which is comparable to the gain accuracy of the AD630. To increase the reference frequency, divide the digital clock’s frequency to select low values when the reference time becomes too long.

See Why JLCPCB Is So Popular: https://jlcpcb.com

You may have to register before you can post comments and get full access to forum. |

Related publications

- Circuits Four-quadrant power supply provides any-polarity voltage and current
- News Analog Devices' Wideband GaN MMIC Amplifier Offers High Gain, Power and Efficiency in Compact Design — HMC8205
- Circuits Amplifier IC Generates a Negative Voltage Reference with the Fewest Parts and a Single Supply Rail
- Circuits Electronic safety-lock with telecard as key
- Datasheets Datasheet AD633ANZ - Analog Devices ANALOG MULITPLIER, 4 QUADRANT, PDIP