Numerous applications require the measurement of very low resistances, including but not limited to fuse integrity analysis, relay characterization, and superconductor evaluation. There is a wide range of commercially available equipment designed for this task, but these units are prohibitively expensive and cannot be practically integrated into many applications outside the laboratory. A common low-cost and compact method of measuring resistance is to inject a known dc current into the unknown resistor, measure the resulting voltage and calculate the resistance using Ohm’s law. Unfortunately, for very low resistances, the current required to generate a voltage sufficiently distinguishable from the surrounding noise becomes impractically large.
|Figure 1.||The AD8253 is configured in transparent gain mode, allowing the gain to be manually set by the user according to Table 1.|
Figure 1 is a schematic of a complete, low-cost, software-programmable lock-in amplifier designed to measure resistances in the micro-Ohm range. An ac voltage is generated by switching current into the unknown resistance. The 1 mA current pulse is generated by using an analog switch (ADG436) to alternately apply 10 V (from the ADR01 precision reference) and ground to the series connection of 10 kΩ and the unknown resistance. The resulting ac voltage, which is imperceptibly small, is then applied to an AD8228 instrumentation amplifier configured in a gain of 100. The low noise of the AD8228 make it well-suited for a first gain stage. Following the AD8228 is an AD8253 software-programmable gain instrumentation amplifier which can be configured in gains of 1, 10, 100, or 1000 depending on the size of the resistance being measured. In Figure 1, the AD8253 is configured in transparent gain mode, allowing the gain to be manually set by the user according to Table 1. The system gain is extremely stable over temperature due to the low gain drift of the AD8228 and AD8253. To extract its amplitude information, a balanced demodulator (AD630) synchronously rectifies the amplified signal. An active three-pole low-pass filter using an AD820 then rejects all out-of-band frequencies, while passing the in-band signal of interest with a transfer function of 50 mV/mΩ at maximum system gain.
|Table 1.||Logic levels for AD8253 transparent|
The system is completely self-contained, and the user only needs to provide a ±15 V supply. A 1 kHz oscillator, constructed from common low-cost digital ICs, creates the CLK signal that controls the excitation and demodulation sections of the system, where the 5 V supply for these ICs is obtained from the 15 V rail using an ADR5045 shunt regulator.
|Figure 2.||The output voltage of the system for various lengths of 18 AWG copper
wire has a resistance of 213.58 μΩ/cm, as measured in the laboratory
The output voltage of the system is linear over the entire range as can be seen in Figure 2.
|Figure 3.||The measurement error of the system, where the noise in the curve
is due to the human error in placing the measurement leads, and the
slope of the curve is due to a slight system gain error.
Figure 3 illustrates the measurement error of the system. The linearity of the system is further demonstrated by Figure 4.
|Figure 4.||The linearity of the system is demonstrated by the output voltage of
the system as the measurement leads are placed across 1, 2, 4, 8,
and 16 cm lengths of wire.
Materials on the topic
- Datasheet Analog Devices AD620
- Datasheet Analog Devices AD820AR
- Datasheet Analog Devices AD8228ARZ
- Datasheet Analog Devices AD8253
- Datasheet Analog Devices ADG436
- Datasheet Analog Devices ADR01
- Datasheet Analog Devices ADR5045
- Datasheet Texas Instruments SN74HC04
- Datasheet Texas Instruments SN74HC74